SECTION I - GETTING STARTED
The DF2000 Digital Indicator is an easy to use, fully featured digital
weight indicator. This manual will lead you through all of the steps required
to set up and operate the indicator.
If you are setting up the indicator for the first time, please refer to
Sections II and III - Installation and Calibration for further instructions.
The DF2000 features three channels and has a maximum displayed
resolution of 5,000 graduations per channel. Each channel is calibrated and
set-up separately and can be calibrated for different graduations sizes and
different capacities from other channels. The indicator can also be operated
in the Total mode if the graduation sizes of the channels are the same. If you
are using less than three channels, the unused channel(s) must be disabled
in order to obtain a correct reading while in the Total mode.
The indicator features a full set of error messages to assist you in
troubleshooting problems during operation and calibration. These messages
are detailed in Section V.